Please note that in case of posterior chamber implantation a different a-constant should be used for the lens calculation. An approximate a-constant can be found on the company website and is also provided on the label on the outer package. This is a theoretical value and should be personalized by the physician.
- Prior to implantation, all packaging should be carefully examined to verify that the correct lens power has been chosen, the expiration date is respected and that the contents have not been damaged. Using a sterile technique, open the blister pack and deliver the lens tray with the lens to the sterile prep tray.
- Make a main incision of 5.5 mm. The main incision should be at 90 degrees from the intended enclavation axis.
- Do NOT inject a miotic agent. Leave the pupil at a minimum size of approximately 3 mm to allow the lens to reach the retropupillary position. Constriction of the pupil will impede the movement of the IOL through the
pupil and might result in damage to the iris tissue.
- Inject a sufficient amount of high viscosity viscoelastic to maintain anterior and posterior chamber depth, but do not overfill. Only use high viscosity sodium hyaluronate viscoelastics such as ArtiVisc® or ArtiVisc Plus®.
- Make a paracentesis of 1.2 mm on either side of the intended main incision site. The paracenteses must be in line with the haptics and pointing slightly downwards.
- Prior to implantation, verify that the correct product and lens power has been selected for the correct eye. The IOL can become electrostatically charged and stick to the lid of the lens tray. Tap slightly on the lid before
opening the lens tray. Open the lens container and examine the lens carefully for damage or debris.
- The IOL may be rinsed with sterile balanced salt solution and coated with a viscoelastic prior to implantation to facilitate easy passing through the main incision.
- Invert the IOL before insertion into the eye.
- Introduce the IOL into the anterior chamber through the main incision in a vertical orientation using the ARTISAN® Implantation forceps. Retract the forceps through the main incision. Care must be taken to avoid
contact with the corneal endothelium.
- Rotate the lens 90 degrees and align with the enclavation axis using the ARTISAN® Lens Manipulator. Center the lens over the pupil.
- Bring the appropriate implantation forceps through the main incision, firmly grasp the IOL at the center of
the optic with the Implantation Forceps and carefully move the IOL through the non-constricted pupil.
- Move the IOL upwards against the iris to facilitate enclavation and choose a side of the lens to enclavate.
- Insert a cannula through one of the paracenteses to fixate the lens to the iris. While securely holding the optic with the implantation forceps, use a cannula to carefully push sufficient mid-peripheral iris tissue, which is virtually immobile, through the slot of the haptics. Sufficient iris tissue must be placed through the haptic slot to ensure adequate lens stability.
- Repeat step 13 at the other side of the lens. Make sure the lens is well centered and verify the amount of iris tissue which is enclavated.
- Perform an iridotomy or iridectomy outside the periphery of the IOL, preferably in the part of the iris that will be covered by the upper eyelid. To prevent light complaints, a single small iridotomy is preferred. For all iris types, the iridotomy is sufficient in size as soon as a red light reflex can be observed. Alternatively, iridotomy may be performed at least 1 week prior to the IOL implantation using a neodymium-doped yttrium aluminium garnet (Nd:YAG) laser.
- Remove all viscoelastic from the eye by flushing with saline using a syringe with cannula. During flushing make a semi-circular movement from opposite the main incision. Also flush out viscoelastic from underneath the IOL. Make sure no viscoelastic remains in the eye as this may cause high intraocular pressure postoperatively.
- Close the main incision to prevent wound leakage. Watertight wound closure is of paramount importance to prevent a shallow anterior chamber in the immediate postoperative period. Do not suture too tight to avoid surgically induced astigmatism. When sutures are made, corneal sphericity can be checked with placido disk under the microscope and, if necessary, adjustments to sutures can be made.
- Administer and prescribe postoperative medication (e.g. antibiotics and corticosteroids).
- A protective patch should be placed over the eye following surgery.
- Patients must be instructed not to rub the eye, to avoid physical impact or direct pressure to the eye and to avoid activities that increase the risk for ocular trauma (e.g. certain martial arts sports) or to wear safety glasses during such activities.
- Patients must be informed that besides a postoperative follow-up after six months, yearly examinations, including endothelial cell counts, are required to assess the long-term safety of the IOL. Furthermore, the p hysician must ensure traceability of the IOL and provide the patient with a patient card.
OPHTEC BV P.O. Box 398 | 9700 AJ Groningen | Schweitzerlaan 15 | 9728 NR Groningen | The Netherlands | T: +31 50 5251944 | F: +31 50 5254386 E: firstname.lastname@example.org | Information on this document is subject to modification | ©2018 OPHTEC BV | L23-HV06-6A-2